5017D, 500mW - 500W Avg. 1000W Peak Wideband Power Sensor
500mW - 500W Avg, 1000W Peak Power
5017D, RF Power Sensor never requires field calibration, only requires calibration once per year and is traceable to National Institute of Standards and Technology (NIST).
The 5017D sensor measures True Average Power, Peak Power, and Duty Cycle directly with exceptional accuracy and uses these precise measurements to calculate a wide range of other important factors, such as VSWR, Return Loss, Reflection Coefficient, Crest Factor, Average Burst Power, and CCDF.
Analog Cellular, Digital Cellular, 3G, 4G, Tetra, DMR, MOTOTRBO, APCO/P25 Phase 1 & 2, Trunking, CDMA, TDMA, WCDMA, GSM, Transportation, Tactical Military, Radar, Avionics, Marine, LMR, Analog Broadcast, Digital Broadcast, GSM, GPRS, EDGE, UMTS, HSDPA, Bluetooth, Fire, GPS, NPSPAC, Paging, Public Safety, Telematics, Utilites, WIMAX and WLAN.
- Frequency Range: 25 MHz - 1.0 GHz
- Power Range: 500mW - 500 Watts Avg, 1000 Watts Peak
( Derate maximum average power rating from 500W at 300MHz to 100W at 1GHZ using a straight line on a log-log scale)
- Insertion VSWR: <1.05
- Insertion Loss: <0.05 dB
- Directivity: 28 dB up to 100 MHz, 30 dB from 100 to 1000 MHz
- Impedance: 50 Ohms (nominal)
- Power Supply: USB Port: Less than one low-power
USB load DC Input Connector: 7-18 VDC at less than 0.1A
- DPM: DB9 proprietary interface
- PC Interface (1): RS-232, 9600 Baud, no parity, 8 data bits, 1 stop bit, DB9
- PC Interface (2): USB 2.0 Type B
- Compliance: CE
Included Accessories: User Manual, VPM3 - Virtual Power Meter Software and Manual, 6 ft. USB Cable.